Development of a high-field MR-guided HIFU setup for thermal and mechanical ablation methods in small animals
- Martijn Hoogenboom1Email author,
- Martinus J. van Amerongen†1,
- Dylan C. Eikelenboom†2,
- Melissa Wassink2,
- Martijn H. den Brok2,
- Christina Hulsbergen-van de Kaa3,
- Erik Dumont5,
- Gosse J. Adema2,
- Arend Heerschap1 and
- Jurgen J. Fütterer1, 4
© Hoogenboom et al. 2015
Received: 5 April 2015
Accepted: 7 August 2015
Published: 13 August 2015
Thermal and mechanical high intensity focused ultrasound (HIFU) ablation techniques are in development for non-invasive treatment of cancer. However, knowledge of in vivo histopathologic and immunologic reactions after HIFU ablation is still limited. This study aims to create a setup for evaluation of different HIFU ablation methods in mouse tumors using high-field magnetic resonance (MR) guidance. An optimized MR-guided-HIFU setup could be used to increase knowledge of the different pathologic and immunologic reactions to different HIFU ablation methods.
Three different HIFU treatment strategies were applied in mouse melanomas (B16): a thermal (continuous wave), a mechanical (5 ms pulsed wave), and an intermediate setting (20 ms pulsed wave) for HIFU ablation, all under MR guidance using a 7 tesla animal MR system. Histopathologic evaluation was performed 3 days after treatment.
The focus of the ultrasound transducer could accurately be positioned within the tumor under MR image guidance, without substantial damage to the surrounding tissue and skin. All mice retained complete use of the treated leg after treatment. Temperatures of >60, <50, and <44 °C were reached during thermal, intermediate, and mechanical HIFU ablation, respectively. Thermal-treated tumors showed large regions of coagulative necrosis. Tumors of both the mechanical and intermediate groups showed fractionated tissue with islands of necrosis and some pseudocysts with hemorrhage.
A stable small animal MR-guided HIFU setup was designed and evaluated for follow-up MR imaging and histopathologic responses of the treated tumors. This will facilitate further studies with a larger number of mice for detailed evaluation of the pathologic and immunologic response to different HIFU strategies.
During the last decades, new treatment options have been developed to treat localized tumors while sparing surrounding tissue. High intensity focused ultrasound (HIFU) is a non-invasive ablation technique to deliver large amounts of energy into a small millimeter-sized ablation zone, resulting in tissue destruction through cellular disruption and irreversible coagulation necrosis [1–3]. If correctly targeted, the energy of the ultrasound (US) beam penetrates the body without damaging the surrounding tissue. Using multiple focal ablations, different tumor volumes can be treated.
Both thermal and mechanical effects can be generated using HIFU . The thermal effect is the consequence of absorption of ultrasound energy by the tissue and conversion into heat, inducing irreversible damage and local coagulative necrosis [4, 5]. The mechanical effect is created by the mechanical force and shock wave effect of the ultrasound wave which could result in acoustic cavitations . With a low duty cycle, the local temperature rise remains limited due to thermal diffusion. Expansion and rarefaction of tissue due to high acoustic waves of a sufficient pressure magnitude result in alternating compression and expansion of the tissue. Subsequently, gas is extracted from the tissue and (micro)bubbles are created. The interaction of these bubbles with the ultrasound waves will result in disruption of the vascular structure [7, 8], connective tissue , and cellular damage. Extreme high ultrasound pressure can destroy the tissue in submicron fragments, so-called histotripsy [10–12] or boiling histotripsy (also known as millisecond boiling) [13–16].
While thermal HIFU ablation is already clinically used for more than a decade, mechanical HIFU ablation techniques are still in an early stage of development [6, 17]. Whether thermal or mechanical HIFU is the best possible way to treat tumors is not known. Moreover, the different pathological and immunological effects due to in vivo thermal or mechanical HIFU treatment are poorly understood. Immune effects have been observed as a response to the HIFU treatment, but further studies are needed to distinguish wound healing from relevant anti-tumor responses [9, 17, 18]. For evaluation of different tumor ablation techniques and their histopathologic changes and immunologic reactions, an adequate small animal HIFU system is required [19, 20], which can be used to test in vivo HIFU treatments .
HIFU guided by US or magnetic resonance imaging (MRI) is widely used . The latter technique offers better anatomical information compared to US-guided techniques, better tumor detection, and a broader field of view to visualize both the treatment area and surrounding tissue . Different MRI methods allow accurate targeting before treatment, real time temperature monitoring, and control of the energy deposited during treatment [1, 2, 21, 23, 24].
The purpose of this feasibility study was to create a controllable setup for evaluation and differentiation between different magnetic resonance (MR)-guided HIFU ablation methods in vivo. Three different methods are tested: (1) thermal (T-HIFU), (2) mechanical (M-HIFU), and (3) a mixed ablation method, termed intermediate temperature-HIFU (inT-HIFU) that resulted in a limited but significant temperature rise in between both other HIFU settings. Tumor response after the different ablation techniques is evaluated with T2-weighted (T2W) MR imaging immediately after treatment. Histopathologic evaluation is performed 3 days after treatment.
Materials and methods
Nine- to eleven-week-old female C57Bl/6n wild type mice were purchased from Charles River Wiga (Sulzfeld, Germany) and kept under specific pathogen-free conditions in the Central Animal Laboratory of the Radboud University (Nijmegen, the Netherlands). All animal experiments were performed according to the guidelines and by approval of the Nijmegen Animal Experiments Committee.
Twenty-one mice were injected subcutaneously in the right leg with 0.5 × 106 cells of the ovalbumin (OVA)-transfected murine melanoma cell line (B16F10-OVA) as previously described . The volume of the tumors was calculated with the formula (A × B2) × 0.4 in which A is the largest and B is the shortest dimension, measured using a caliper. Eight to nine days after injection, a tumor diameter of more than 7 mm is reached and the mouse was ready for HIFU treatment.
Isoflurane gas at 3.5 % was used for anesthetic induction. The anesthetic concentration was adjusted to 1–2 % during the experiment in order to maintain the breathing frequency at 40–60 per minute. The body temperature was measured during the treatment using a rectal thermometer and maintained using a heated air flow device. The mice were euthanized 3 days after treatment by cervical dislocation.
Animal setup and protocol
An MR compatible animal HIFU system (Image Guided Therapy—IGT, Pessac, France) was used in all experiments. A 16-channel annular array HIFU transducer (3 MHz central frequency, 37 W acoustic peak power, 52 W electrical peak power, 86.58° aperture, 48 mm diameter, adjustable focus depth 30–80 mm) was embedded in a positioning system (MR compatible piezoelectric motors, 30 × 30 mm trajectory execution range, active transducer cooling system) which was controlled by trajectory planner software (Thermoguide, IGT, Pessac, France). The spherical cap of the transducer is divided into 16 concentric annuli of identical surface. Due to the cylindrical symmetry, the steering of the ultrasound focus is possible along the axis of the symmetry i.e., longitudinal positioning of the focal zone. Lateral positioning of the focal zone is processed due to transducer displacement via piezoelectric motors. The trajectory of ablation was manually implemented based on MR imaging.
The pressure field of the HIFU transducer was scanned using a 0.075 mm needle-type hydrophone (HPM075/1, Precision Acoustics, Dorchester, UK) and 50 μs pulses with an acoustical output power of 0.73 W (electrical output power of 1.04 W). The measurements were performed in a degassed water tank. The hydrophone was positioned into a 3D mechanical positioning system enabling a 0.05 mm measurement resolution.
To visualize the border between treated and non-treated tumor tissue, three to six focused spots were positioned in the center of the tumor depending on tumor size and treatment method used. Thermal diffusion, which occurs in thermal HIFU treatment, results in an increased area of tumor destruction of each ablation compared to the M-HIFU or inT-HIFU treatment. Therefore, three to four focal sonications, 2 mm apart, were used during thermal treatment (T-HIFU) and a maximum of six focal sonications (1.5 mm apart) for M-HIFU and inT-HIFU treatment. Six mice were subjected to T-HIFU with a continuous wave HIFU pulse of 4 s per sonication. Five mice were treated with the inT-HIFU method, 120 pulses of 20 ms (inT-HIFU treatment) with a pulse repetition frequency of 4 Hz. Six mice received a more mechanical HIFU ablation method (M-HIFU) with 500 pulses of 5 ms with a pulse repetition frequency of 4 Hz. For all methods, T-HIFU, M-HIFU, and inT-HIFU, an acoustic output power of 40–48 W was used, which correlates approximately to an acoustic dose per sonication of 160, 100, and 96 J. One mouse was excluded due to bursting of the tumor before treatment was feasible. Three tumor-bearing mice did not receive any treatment and were used as a control group.
The HIFU settings were directed by the temperature rise during ablation and the corresponding response in phantoms and ex vivo experiments. The T-HIFU group received a continuous US wave causing the temperature to rise within the focal region. Ex vivo experiments showed, with these settings, an increase of a minimum of 30 °C. The pulsed waved character of M-HIFU and inT-HIFU methods was used to limit the temperature rise and give a more mechanical character to the treatment method. The same pulse repetition frequency (4 Hz), but a different pulse duration and amount of pulses, was chosen for the M-HIFU and inT-HIFU methods (500 pulses of 5 ms and 120 pulses of 20 ms, respectively). The HIFU on time was similar, i.e., 2400 and 2500 ms, and the temperature increased in ex vivo specimens was 8 and 15 °C.
MR imaging and thermometry
For MR imaging and treatment guidance, a 7 tesla horizontal wide bore (20 cm) animal MR scanner was used (ClinScan, Bruker Biospin GmbH, Rheinstetten, Germany), with an additional body coil insert (free bore diameter 15 cm). Before treatment, an axial susceptibility sensitive T1 weighted (T1W) gradient echo (GRE) sequence (repetition time/echo time (TR/TE) of 196/4 ms, flip angle (FA) of 250, voxel size 0.51 × 0.51 × 2.50 mm) was acquired to check for the presence of air bubbles within the line of the acoustic beam (Fig. 2a). Mice were repositioned if gas bubbles were present. Subsequently, localized T1W MR images were acquired with a gradient echo fast low angle shot (GRE-FLASH) sequence (TR/TE = 236/4 ms, FA = 250, voxel size 0.78 × 0.78 × 1.50 mm) and transferred to the HIFU workstation for synchronization of the coordinates of the HIFU transducer with the coordinates of the MR scanner. Before and after the HIFU treatment, anatomical T2W MR images were acquired for tumor localization (Fig. 2b), trajectory planning, and tumor response analysis, using a turbo spin echo (TSE) scan (TR/TE = 3350/26 ms, FA = 1800, voxel size 0.25 × 0.25 × 1 mm, TSE-factor 7). The T2W images were sent to the HIFU trajectory planning software (Thermoguide, IGT).
Before the in vivo mouse experiments, the stability and accuracy of the MR thermometry sequences were determined within chicken filet and a 3 % agarose phantom. The stability of the MR thermometry was measured in a chicken filet and in a mouse. The temperature within the chicken filet without heating was measured for 2 min. The accuracy of the MR thermometry was assessed with a 3 % agarose phantom. The agarose phantom was positioned in boiled water to increase the temperature. The temperature changes were measured with MR thermometry. Within both experiments, a fiber optic temperature probe (T1S-2 M probe, Neoptix, Québec, Canada) was positioned within the phantom or chicken filet as a standard of reference. Within the mouse, a rectal thermometer was used as standard of reference.
Mice were sacrificed 3 days after HIFU treatment. The tumors were removed and inked for visual orientation after which they were fixated in formalin. The tumors were cut in half, parallel to the surface of the epidermis using the same orientation as the T2W MR imaging made after treatment, under visual inspection. After paraffin embedding, 4-μm thick tissue sections were stained with hematoxylin and eosin (H&E). Immunohistochemical staining by F4/80 was used to investigate macrophage infiltration in the tumor. Paraffin tissue sections (4 μm) were deparaffinized and subjected to antigen retrieval by citrate (10 mM) with microwave treatment for 10 min. Then the tissue sections were incubated with serum to block aspecific bindings, after which the sections were incubated with F4/80 antibodies (1:20 in 2 % normal goat serum (NGS), eBioscience, Vienna, Austria) at room temperature. The first antibody was coupled with a goat anti-rat-biotin antibody (1:100 in 1 % NGS/1 % normal mouse serum) and colored with Fast Red bound to ABC-Horseradish Peroxidase. The sections were counterstained with hematoxylin.
Therapy effects were evaluated and scored on H&E slides by a pathologist. Morphologic changes were evaluated, such as percentage of necrosis, type of tissue damage (apoptosis versus coagulative necrosis), and type of tumor necrosis (confluent versus fractionated). Furthermore, ballooning degeneration, dilatation of capillaries, cysts, and bleeding were recorded. Finally, lymphocytic infiltration, both within and on the tumor edge, was assessed.
MR thermometry and ex vivo experiments
The hydrophone measurements of the focal region revealed a cigar shaped volume of 0.5 × 0.5 × 2 mm (full width half maximum (FWHM)). At an acoustic output power of 1.04 W, the peak pressure was 1.8 MPa. The gel pad positioned within the US beam did not show any interference with the US beam or a decrease of the focal pressure, data not shown.
The stability and accuracy of the MR thermometry was evaluated by measuring the temperature within a chicken filet, one mouse and an agarose phantom. The temperature stability was assessed in the absence of heating. The standard deviation of the temperature measured in a chicken filet over 2 min was +/− 0.34 °C with the GRE-flash sequence (temporal resolution 3.8 s) and +/−0.16 °C with the EPI sequence (temporal resolution 1.9 s). The temperature accuracy was determined while heating up an agarose phantom. The average deviation between the MR thermometry and the fiber optic temperature probe was −0.21 °C (±0.52) and 1.6 °C (±1.8), respectively, with the GRE-flash and EPI sequences. Within a non-heated area of the mouse tumor MR thermometry with the EPI sequence revealed a variation of +/−0.31 °C with baseline corrections.
The mean tumor dimensions were A = 8.7 mm (±1.5) and B = 7.3 mm (±1.1) on the day of treatment. The mean tumor volume of the mice was 194 mm3, with a range between 111 and 285 mm3, but one mouse had a relatively small (82 mm3) and another a large (364 mm3) tumor in contrast to the other tumors. For T-HIFU or M/inT-HIFU treatment, respectively, three or six sonications were positioned within the tumor except one mouse of the M-HIFU group with a tumor size of 82 mm3, for which only two focused sonications were applied. One mouse was excluded due to unexpected tumor burst before treatment, leaving a total of 17 mice which were treated with HIFU. Six of them in the thermal HIFU group, 5 in the M-HIFU, and 6 mice in the inT HIFU group. Three mice were used as a control. In three mice, tumor growth was restricted by adjacent muscle tissue, which might have influenced the treatment due to difference in absorption rate of the tissue. The ultrasound beam path was clearly visualized in all mice in T1W MR images (Fig. 2a).
The body temperature during treatment of the anesthetized mice ranged between 34 and 37 °C. Two mice had to be repositioned due to air bubbles within the US beam path, which resulted in a drop of the body temperature (30 °C), but the body temperature was stabilized after repositioning, and both mice recovered normally after treatment. In all experiments the focus of the HIFU system was well aligned after one or two test pulses inside the phantom cube. As observed on the MR images and histopathological slices, all ablations were accurately positioned inside the tumor in all mice. No skin burns were observed immediately after the HIFU treatment. All mice (n = 17) retained complete use of the treated leg within 15 min after treatment without showing signals of pain.
Two mice of the T-HIFU group showed a small skin defect covering the tumor 2 and 3 days after treatment. They had some difficulties using the leg with the tumor, possibly due to necrosis of the tumor directly underneath the skin or heating of tissue next to the tumor. Two mice of the inT-HIFU group also revealed small skin defects on top of the tumor. However, they did not show any discomfort and used their legs properly. The size of the tumors of five mice treated with T-HIFU ablation, four tumors of the inT-HIFU-treated mice, and three tumors of the M-HIFU-treated mice diminished in the days after treatment, which was not seen in the control mice.
Both the M-HIFU and inT-HIFU group revealed less coagulative necrosis, and more fractionated necrosis and apoptotic cells, compared to one big necrotic region seen in the T-HIFU-treated group. These tumors showed islands of necrosis within vital tumor cell areas. The total necrosis within the treated area was 26 % (sd. 11 %) in the inT-HIFU group and 22 % (sd. 9 %) in the M-HIFU-treated mice. Histopathologically, the H&E sections revealed similar results between the M-HIFU or inT-HIFU ablation settings. In both groups, some hemorrhage and ballooning degeneration was observed. In two mice of the inT-HIFU-treated mice, small pseudocysts within the tumor were found, most of which were filled with erythrocytes (Fig. 6e–h).
All tumors showed moderate lymphocyte infiltration surrounding the tumor. The M-HIFU and inT-HIFU treated mice also showed additional lymphocyte infiltration within the center of the tumor, without differences between the two other groups. In all mice, the F4/80 macrophage staining detected macrophages in the immediate surroundings of the tumor and sometimes near fragmented necrotic regions. However, no quantitative differences were found between the three treatment methods at the time points analyzed.
A new MR-guided HIFU setup for the treatment of small animals was designed and evaluated with three different ablation methods. This setup can be used to distinguish between thermal and mechanical ablation methods. The small focus (0.5 × 0.5 × 2.0 mm FWHM) of the HIFU system results in a high ablation accuracy. The HIFU focus can accurately be positioned within the tumor with minimal to no damage to the surrounding tissue, such as the skin or leg muscles. In combination with the high-field small animal MR system, the temperature within the focal region and surrounding tissue can accurately (+/−0.31 °C) be measured with a high spatial resolution in real time (temporal resolution 1.9 s). Thus, it is now feasible to do larger follow-up studies in mice after MR-guided HIFU treatment to improve treatment methods and monitoring of tumor response.
Two different thermometry methods were used, a GRE-FLASH sequence and an EPI sequence. With the GRE-FLASH sequence, the temperature can only be visualized after each sonication instead of real time visualization due to technical limitations of the sequence. Using the EPI sequence, a thermal map can be displayed immediately during the ablation. The GRE-FLASH sequence created accurate thermometry maps (temperature uncertainty <0.52 °C) and anatomical imaging with a relatively high spatial resolution (0.78 × 0.78 × 1.5 mm3, five slices), each 3.8 s. This allowed controllable HIFU and proper follow-up. However, due to the interleaved ablation and visualization of the thermal map, occasionally heating of the surrounding tissue was noticed after ablation, which suggests that direct visualization is required to observe heating of surrounding tissue on time. Furthermore, the fast temperature rise combined with a relatively low temporal resolution (3.8 s) might result in under-sampled and inaccurate temperature measurements due to partial volume effects. With the real time EPI sequence, a less accurate thermometry map was obtained (uncertainty <1.8 °C), in comparison with the GRE-FLASH sequence, and a decreased contrast of the images, making tumor border detection more difficult. However, due to the fast imaging (1.9 s per dynamic per 4 slices), less under-sampling occurred. Also a slice perpendicular to the transversal slices was used to visualize the tissue above the focus of the US beam. Due to the direct visualization of the tumor and surrounded tissue, immediate abortion of the ablation was feasible and any heating of surrounding tissue was noticed immediately. Optimization of the real time imaging is required for better visualization of the tumor margins. While the focal region was easily visualized with T-HIFU or inT-HIFU ablation, during M-HIFU treatment, the temperature change was limited (<7 °C increase) and little or no signal changes were observed in the images during treatment. This results in difficulties to confirm the location of the focus in real time and to determine the treatment size created by M-HIFU treatment. Possibilities to visualize the focal zone with mechanical HIFU treatment might be MR acoustic radiation force imaging .
Only in three out of six T-HIFU-treated mice, high intensity focal spots were seen in T2W MR images after treatment, while all mice showed histopathologic changes. This might be because T2W MR images were acquired immediately after treatment. Tissue response might become more apparent some hours after treatment . Next to T2W MR imaging, other MR imaging techniques such as (dynamic) contrast-enhanced MR imaging, diffusion weighted, or T2* weighted MR imaging could be used for treatment evaluation and follow-up [21, 30–33]. It has previously been shown that the permeability of vessels increases, or the vessels are destroyed by mechanical destruction for example due to cavitation effects . The permeability decreases due to tissue necrosis by thermal destruction of tissue, which results in an increase and decrease of contrast uptake . Therefore, contrast MR imaging could visualize both thermal and mechanical HIFU treatment response and to help distinguish the thermal or mechanical bioeffects of different ablation methods.
Although no significant changes were observed in the MR images directly after treatment, microscopic changes due to treatment were found in all tumors. Instead of treating the entire tumor, only a small amount (3–6) of focal ablations was positioned inside the tumor. Histopathologically, the thermal-treated mice revealed more confluent coagulation necrosis, while both M-HIFU- and inT-HIFU-treated groups showed more fractionated tissue necrosis and hemorrhage. These effects are similar as found in earlier studies on sheep, rats, or rabbits [8, 35, 36]. The mechanical effects were limited compared to thermal effects, still some dying cells and living cells were found between the fractionated tissues 3 days after the ablation. Real cavitation effects were not found, possibly because only transient damage was created while the tumor was removed after 3 days. Further, increasing the acoustical output power might be a solution for a more complete mechanical destruction of the tissue, as is shown in boiling histotripsy ablation methods [6, 14]. In future studies, a longer follow-up after treatment is of interest to evaluate further tumor responses after the different ablation techniques.
In this study, a stable MR guided HIFU treatment setup is designed to treat mice using either T-HIFU, M-HIFU, or an inT-HIFU approach. It is shown that HIFU ablation of murine melanoma tumors is feasible and causes different pathologic effects within the tumor with high accuracy, without damaging the surrounding tissue and with little or no damage to the skin. This design allows to improve and to investigate different HIFU treatment methods and their effects in vivo for larger cohort of animals.
This work was supported by a RadboudUMC PhD grant and the Center for Translational Molecular Medicine (VOLTA).
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Jolesz FA. MRI-guided focused ultrasound surgery. Annual review of medicine. 2009;60:417–30.Google Scholar
- Petrusca L, Salomir R, Milleret R, Pichot O, Rata M, Cotton F, et al. Experimental investigation of thermal effects in HIFU-based external valvuloplasty with a non-spherical transducer, using high-resolution MR thermometry. Phys Med Biol. 2009;54:5123–38.Google Scholar
- Haar GT, Coussios C. High intensity focused ultrasound: physical principles and devices. International journal of hyperthermia : the official journal of European Society for Hyperthermic Oncology, North American Hyperthermia Group. 2007;23:89–104.Google Scholar
- Dubinsky TJ, Cuevas C, Dighe MK, Kolokythas O, Hwang JH. High-intensity focused ultrasound: current potential and oncologic applications. AJR American journal of roentgenology. 2008;190:191–9.Google Scholar
- Sapareto SA, Dewey WC. Thermal dose determination in cancer therapy. International journal of radiation oncology, biology, physics. 1984;10:787–800.Google Scholar
- Hoogenboom M, Eikelenboom D, den Brok MH, Heerschap A, Futterer JJ, Adema GJ. Mechanical High-Intensity Focused Ultrasound Destruction of Soft Tissue: Working Mechanisms and Physiologic Effects. Ultrasound in medicine & biology. 2015.Google Scholar
- Liu HL, Wai YY, Chen WS, Chen JC, Hsu PH, Wu XY, et al. Hemorrhage detection during focused-ultrasound induced blood-brain-barrier opening by using susceptibility-weighted magnetic resonance imaging. Ultrasound in medicine & biology. 2008;34:598–606.Google Scholar
- Chen H, Brayman AA, Bailey MR, Matula TJ. Blood vessel rupture by cavitation. Urological research. 2010;38:321–6.Google Scholar
- Liu F, Hu Z, Qiu L, Hui C, Li C, Zhong P, et al. Boosting high-intensity focused ultrasound-induced anti-tumor immunity using a sparse-scan strategy that can more effectively promote dendritic cell maturation. J Transl Med. 2010;8:7.Google Scholar
- Xu Z, Raghavan M, Hall TL, Mycek MA, Fowlkes JB. Evolution of bubble clouds induced by pulsed cavitational ultrasound therapy - histotripsy. IEEE Trans Ultrason Ferroelectr Freq Control. 2008;55:1122–32.Google Scholar
- Xu Z, Fan Z, Hall TL, Winterroth F, Fowlkes JB, Cain CA. Size measurement of tissue debris particles generated from pulsed ultrasound cavitational therapy-histotripsy. Ultrasound in medicine & biology. 2009;35:245–55.Google Scholar
- Roberts WW, Hall TL, Ives K, Wolf JS, Jr., Fowlkes JB, Cain CA. Pulsed cavitational ultrasound: a noninvasive technology for controlled tissue ablation (histotripsy) in the rabbit kidney. The Journal of urology. 2006;175:734–8.Google Scholar
- Simon JC, Sapozhnikov OA, Khokhlova VA, Wang YN, Crum LA, Bailey MR. Ultrasonic atomization of tissue and its role in tissue fractionation by high intensity focused ultrasound. Physics in medicine and biology. 2012;57:8061–78.Google Scholar
- Khokhlova TD, Canney MS, Khokhlova VA, Sapozhnikov OA, Crum LA, Bailey MR. Controlled tissue emulsification produced by high intensity focused ultrasound shock waves and millisecond boiling. The Journal of the Acoustical Society of America. 2011;130:3498–510.Google Scholar
- Wang YN, Khokhlova T, Bailey M, Hwang JH, Khokhlova V. Histological and biochemical analysis of mechanical and thermal bioeffects in boiling histotripsy lesions induced by high intensity focused ultrasound. Ultrasound in medicine & biology. 2013;39:424–38.Google Scholar
- Khokhlova VA, Bailey MR, Reed JA, Cunitz BW, Kaczkowski PJ, Crum LA. Effects of nonlinear propagation, cavitation, and boiling in lesion formation by high intensity focused ultrasound in a gel phantom. J Acoust Soc Am. 2006;119:1834–48.Google Scholar
- Hu Z, Yang XY, Liu Y, Sankin GN, Pua EC, Morse MA, et al. Investigation of HIFU-induced anti-tumor immunity in a murine tumor model. Journal of translational medicine. 2007;5:34.Google Scholar
- Hu Z, Yang XY, Liu Y, Morse MA, Lyerly HK, Clay TM, et al. Release of endogenous danger signals from HIFU-treated tumor cells and their stimulatory effects on APCs. Biochem Biophys Res Commun. 2005;335:124–31.Google Scholar
- den Brok MH, Sutmuller RP, van der Voort R, Bennink EJ, Figdor CG, Ruers TJ, et al. In situ tumor ablation creates an antigen source for the generation of antitumor immunity. Cancer research. 2004;64:4024–9.Google Scholar
- den Brok MH, Sutmuller RP, Nierkens S, Bennink EJ, Frielink C, Toonen LW, et al. Efficient loading of dendritic cells following cryo and radiofrequency ablation in combination with immune modulation induces anti-tumour immunity. British journal of cancer. 2006;95:896–905.Google Scholar
- Hijnen NM, Heijman E, Kohler MO, Ylihautala M, Ehnholm GJ, Simonetti AW, et al. Tumour hyperthermia and ablation in rats using a clinical MR-HIFU system equipped with a dedicated small animal set-up. Int J Hyperthermia. 2012;28:141–55.Google Scholar
- Tempany CM, McDannold NJ, Hynynen K, Jolesz FA. Focused ultrasound surgery in oncology: overview and principles. Radiology. 2011;259:39-56.Google Scholar
- Rieke V, Butts Pauly K. MR thermometry. Journal of magnetic resonance imaging : JMRI. 2008;27:376–90.Google Scholar
- Ishihara Y, Calderon A, Watanabe H, Okamoto K, Suzuki Y, Kuroda K, et al. A precise and fast temperature mapping using water proton chemical shift. Magnetic resonance in medicine : official journal of the Society of Magnetic Resonance in Medicine / Society of Magnetic Resonance in Medicine. 1995;34:814–23.Google Scholar
- McDannold N. Quantitative MRI-based temperature mapping based on the proton resonant frequency shift: review of validation studies. International journal of hyperthermia : the official journal of European Society for Hyperthermic Oncology, North American Hyperthermia Group. 2005;21:533–46.Google Scholar
- Yuan J, Mei CS, Panych LP, McDannold NJ, Madore B. Towards fast and accurate temperature mapping with proton resonance frequency-based MR thermometry. Quantitative imaging in medicine and surgery. 2012;2:21-32.Google Scholar
- El-Sharkawy AM, Schar M, Bottomley PA, Atalar E. Monitoring and correcting spatio-temporal variations of the MR scanner's static magnetic field. Magma. 2006;19:223–36.Google Scholar
- Auboiroux V, Viallon M, Roland J, Hyacinthe JN, Petrusca L, Morel DR, et al. ARFI-prepared MRgHIFU in liver: simultaneous mapping of ARFI-displacement and temperature elevation, using a fast GRE-EPI sequence. Magnetic resonance in medicine : official journal of the Society of Magnetic Resonance in Medicine / Society of Magnetic Resonance in Medicine. 2012;68:932–46.Google Scholar
- Hectors SJ, Jacobs I, Strijkers GJ, Nicolay K. Multiparametric MRI analysis for the identification of high intensity focused ultrasound-treated tumor tissue. PloS one. 2014;9:e99936.Google Scholar
- Kim YS, Lim HK, Kim JH, Rhim H, Park BK, Keserci B, et al. Dynamic contrast-enhanced magnetic resonance imaging predicts immediate therapeutic response of magnetic resonance-guided high-intensity focused ultrasound ablation of symptomatic uterine fibroids. Investigative radiology. 2011;46:639–47.Google Scholar
- Wijlemans JW, Deckers R, van den Bosch MA, Seinstra BA, van Stralen M, van Diest PJ, et al. Evolution of the ablation region after magnetic resonance-guided high-intensity focused ultrasound ablation in a Vx2 tumor model. Investigative radiology. 2013;48:381–6.Google Scholar
- Liao AH, Liu HL, Su CH, Hua MY, Yang HW, Weng YT, et al. Paramagnetic perfluorocarbon-filled albumin-(Gd-DTPA) microbubbles for the induction of focused-ultrasound-induced blood-brain barrier opening and concurrent MR and ultrasound imaging. Physics in medicine and biology. 2012;57:2787–802.Google Scholar
- Jacobs MA, Gultekin DH, Kim HS. Comparison between diffusion-weighted imaging, T2-weighted, and postcontrast T1-weighted imaging after MR-guided, high intensity, focused ultrasound treatment of uterine leiomyomata: preliminary results. Medical physics. 2010;37:4768–76.Google Scholar
- O’Neill BE, Vo HQ, Shao H, Karmonik C, Zhou X, Li KC. MRI-based prediction of pulsed high-intensity focused ultrasound effect on tissue transport in rabbit muscle. Journal of magnetic resonance imaging : JMRI. 2013.Google Scholar
- Jiang L, Hu B, Guo Q, Chen L. Sonographic and histological development of high-intensity focused ultrasound in rabbit muscle. Experimental and therapeutic medicine. 2013;5:33–8.Google Scholar
- Kim Y, Gelehrter SK, Fifer CG, Lu JC, Owens GE, Berman DR, et al. Non-invasive pulsed cavitational ultrasound for fetal tissue ablation: feasibility study in a fetal sheep model. Ultrasound Obstet Gynecol. 2011;37:450–7.Google Scholar