- Oral presentation
- Open Access
Usefulness of 3D slicer for the planning and monitoring of hepatocellular carcinoma treatment using FUS
© Doba et al; licensee BioMed Central Ltd. 2015
- Published: 30 June 2015
- Coagulation Necrosis
- Optical Tracker
- Poor Visualization
- Synchronous Movement
- Navigation Software
FUS is a noninvasive treatment method, as complete coagulation necrosis is achieved without the insertion of any instruments. However, FUS monitor has the poor visualization because of the presence of the multi-reflections, rib shadows and the emergence of the hyperecho after the FUS treatment. 3D Slicer imaging is a diagnostic imaging support system that can provide the same cross-sectional MPR images on the same monitor screen using DICOM volume data from MRI which are not influenced by those artifacts. The purpose of this study was to utilize an interventional navigation system designed for FUS assisted by 3D Slicer was proposed, and a phantom study was carried out to assess the proposed system.
The FUS system (Mianyang Haifu Tech) was used under ultrasound guidance. In this system, the open-source navigation software is connected together images using an open network communication protocol, OpenIGTLink. A Polaris Vicra optical tracker (Northern Digital, Ontario, Canada) was used. MRI scans (Signa HDX 3.0T system; GE Healthcare) were performed, and 3D Slicer was customized to combine MR images for the navigation.
Testing was performed using an abdominal phantom (CIRS Model057, Norfolk, VA).
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.