Evaluation of MRI protocols for the assessment of lumbar facet joints after MR-guided focused ultrasound treatment
© Krug et al. 2016
Received: 24 January 2016
Accepted: 23 March 2016
Published: 6 April 2016
MR-guided focused ultrasound (MRgFUS) might be a very safe and effective minimally invasive technique to treat facet joint pain caused by arthritis and other degenerative changes. However, there are still safety concerns for this treatment and challenges regarding MR imaging and temperature mapping due to susceptibility effects between the bone and soft tissue near the joint, which has resulted in poor MR image quality. The goal of this research was to evaluate multiple magnetic resonance imaging (MRI) pulse sequences for characterizing ablated lumbar facet joint lesions created by high-intensity focused ultrasound (FUS) and compare the findings to histological tissue assessment. In particular, we investigated the use of T2-weighted MRI to assess treatment effects without contrast administration.
An IACUC approved study (n = 6 pigs) was performed using a 3T widebore MRI system equipped with an MRgFUS system. Facet joints of the lumbar vertebra were ablated using 1-MHz frequency and multiple sonication energies (300–800 J). In addition to T2-weighted MRI for treatment planning, T1-, T2-, and T2*-weighted and perfusion MRI sequences were applied. Signal intensity ratios of the lesions were determined. Histopathology was used to characterize cellular changes.
Ablation of the facet joint, using MRgFUS, was successful in all animals. T2-weighted images showed high signal intensity in the edematous facet joint and adjacent muscle, while delayed contrast-enhanced T1-weighted images showed an enhanced ring surrounding the target volume. T2*-weighted GRE images revealed inconsistent lesion visualization. Histopathology confirmed the presence of cellular coagulation (shrinkage), extracellular expansion (edema), and hemorrhage in the bone marrow.
MRgFUS provided sufficient precision and image quality for visualization and characterization of ablated facet joints directly after ablation. MRI may help in monitoring the efficacy of FUS ablation without contrast after treating patients with back pain.
KeywordsMRgFUS Facet joint T2-weighted FSE Delayed contrast-enhanced MRI
Interest in MR-guided focused ultrasound (MRgFUS) has increased tremendously over the past few years. In MRgFUS, focused ultrasound beams are combined with real-time magnetic resonance imaging (MRI) monitoring to perform controlled thermal ablation . This combined modality is well suited for targeting, characterizing, and quantifying pathologic tissues [2–4]. MRgFUS has recently emerged as an effective treatment option for ablating uterine fibroids [5–10], bone metastases , breast cancer , hepatocellular carcinoma , brain cancer , and treating brain cancer by opening the blood-brain barrier . Recently, MRgFUS has also been applied to the treatment of facet joints in patients with lower back pain and bone metastasis [16, 17]. Significant improvement in patient lifestyle was evident after treatment .
In MRgFUS, image guidance is extremely important for both treatment planning and real-time temperature monitoring. MRI can provide instant information on temperature changes in tissues  that plays an important role in treatment planning and tissue necrosis. It also ensures safety monitoring of the procedure and energy deposition. Although much attention has been paid to the optimization of thermometry approaches for MRgFUS [19, 20], less focus has been placed on the optimization of MR protocols for characterization of ablated tissues after treatment. In particular, imaging techniques for lesion detection without the need of contrast would allow potential repetitions of focused ultrasound (FUS) treatment within the same session. However, this is not possible once contrast has been administered . Hijnen et al.  investigated several effects of contrast agent before FUS treatment. They found a significant frequency shift due to local magnetic field from the contrast agent, which resulted in wrong calculations for temperature and dose. Furthermore, short-term trapping of contrast in the coagulated tissue volume was observed, but no effect on long-term retention was found. Thus, lesion visualization without the need of contrast would be an important progress for clinical applications of MRgFUS. Harnof et al. recently showed that visualization and characterization of facet joint lesions after thermal ablation is particularly challenging with very limited MR image quality . Thus, the goal of this study was to further optimize and evaluate multiple MRI sequences for characterizing ablated lumbar facet joint created by FUS.
All experimental procedures received approval from the Institutional Animal Care and Use Committee (IACUC). Six healthy female farm pigs (Pork Power Farms, Turlock, CA) were premedicated with 0.5 mg/kg acepromazine (PromAce; Fort Dodge Animal Health, Fort Dodge, IA) and 30 min later 25 mg/kg ketamine (Ketaset; Fort Dodge Animal Health). The pigs were then anesthetized with a mixture of isoflurane 2–5 % and oxygen. Saline (10 mL/kg/h) was intravenously infused throughout the experiment for hydration. Vital signs (heart rate, electrocardiogram (ECG), respiratory rate, O2-saturation) were monitored throughout the procedure. To minimize muscle twitching and diaphragm motion during inspiration and expiration in anesthetized pigs, rocuronium (0.1 mg/kg) was intravenously administered prior to ablation.
MRgFUS setup and treatment
All FUS experiments were performed using the ExAblate 2000 System (InSightec Ltd., Tirat Carmel, Israel) with a phased array transducer of 208 elements embedded in the MR scanner table. The table was connected to a 3T MRI widebore scanner (see details below). The skin around the targeted treatment areas was shaved for each animal, cleaned, and closely examined for any skin defects or scars, which might impede the propagation of acoustic energy from the transducer. Each pig was then placed onto a gel pad on the scanner table in a supine position, inside a shallow bath filled with degassed water.
We have treated one side of three to four vertebral levels, thus three to four facet joints per animal. We have identified a central MRI slice from the planning images of each facet joint and added one more slice on each side for the whole treatment volume. We have applied two partially overlapping sonication on each slice resulting in a total of six sonications per joint. The temperature was assessed during each sonication using echo planar imaging (EPI) as previously described [20, 23]. The temperature rise was measured at the focal point and in the far field adjacent to the intervertebral foramen.
MR sequences and imaging parameter
Flip angle (°)
In-plane resolution (mm)
Image and statistical analysis
On all acquired images, the patterns of delineation of the treated facet joints were compared to the untreated contralateral joints. Signal to noise (SNR) was evaluated as the mean signal intensity (SI) in the region of interest divided by the standard deviation of the noise, while SI ratio (SIR) was determined by dividing SI of ablated lesion over surrounding normal tissues. Thus, SIR <1 was defined as hypointense, SIR = 1 as isointense, and SIR >1 as hyperintense. Contrast to noise ratio (CNR) was evaluated by the difference in signal between two regions divided by the standard deviation of the noise. Images were assessed for the presence of edema on the T2-FSE images as well as loss of cellular and vascular integrity on DCE T1-FSE and T2*-GRE images. SI(s) as a function of time were also measured in ablated and surrounding normal tissues on perfusion images. Paired Student’s t test was used for statistical analysis. Data were presented as mean ± SEM and a P value less than 0.05 was considered significant.
Animals were heparinized prior to euthanasia. All animals were sacrificed ~4 h after ablation by IV injection of saturated potassium chloride. Formalin (4 %) was infused for 1 h to fix the tissue in situ. Tissue samples (treated and contralateral non-treated facet joints and paravertebral muscles) were obtained, sliced, and fixed in 70 % formalin. Microscopic sections (5 μm) were stained with hematoxylin and eosin stain (H&E) and examined microscopically. Histopathological sections through the paravertebral muscles, dorsal root ganglion, and facet joint were acquired.
The instant temperature changes were monitored during treatment using the standard proton resonance frequency (PRF) method with echo planar MRI . The mean temperature during sonication was 62.7 ± 9.8 °C. After the procedure, the capability of each pulse sequence to depict the ablated facet joints and adjacent tissue was determined. Table 1 shows the differences in scan time and spatial resolution between the used pulse sequences. Our study showed that the image quality from the body coil was sufficient to clearly visualize the lesions. We found a gain in SNR of a factor of 1.7 by switching the body coil to a 64 channel cardiac coil, but this required changing the table and position of the animal and may lead to misregistration of the images.
A lack of differential contrast between the ablated facet joint or paravertebral muscles and normal adjacent tissues was found on first-pass perfusion images, which can be attributed to a trade-off between time resolution, spatial resolution, number of covered slices, and slice thickness.
Using MRgFUS, the ideal imaging protocol should have the following two characteristics. First, it should be highly sensitive to acute tissue changes to allow for accurate evaluation of the treatment. Second, it should allow repetitions of the treatment if necessary. The later demand would exclude the use of contrast agents. The goal of this work was to present a comprehensive study for MRgFUS treatment of the facet joint investigating several MR pulse sequences, sonication energies, and histological findings. Compared to a previous study by Harnof et al. , we have further enhanced MR imaging using T1-FSE with and without contrast, T2-FSE, T2*-GRE, and perfusion sequences and confirmed outcomes by histology. One of the major findings of our study is that T2-FSE MRI can provide excellent post-FUS lesion detection without the need of contrast administration. This has important implications for the FUS procedure because the ablation can be immediately evaluated and retreated if necessary. This is not possible once contrast has been administered.
Both pre-contrast T2-FSE and post-contrast DCE T1-FSE sequences have the potential to delineate ablated facets and surrounding tissues. Microscopic examination of biopsies confirmed the thermal effects on tissues adjacent to the facet joints and lumbar vertebrae. T2-FSE showed strong positive contrast with hyperintense lesion signal suggesting the presence of edema in the lesion. In addition, slightly increased blood flow (hyperemia) could also be observed in the peripheral zones similar to DCE (Figs. 3, 4, and 5). Although the scan time was slightly increased on T2-FSE scans (Table 1), there was no need for administration of contrast media or delay time (typically 10 min) for imaging. Thus, this might be a fast approach for lesion detection and verification immediately after treatment. We found that a 3-mm slice thickness and 192 matrixes provided the best trade-off between image quality, contrast, and scan time for all sequences (Table 1). Previously, inconsistent results have been reported using T2-weighted images, e.g., heterogeneous appearance was shown after FUS in uterine fibroids , prostate , and pancreatic tumors .
In addition to T2-FSE and DCE, T2*-GRE might be useful to distinguish edema from hemorrhage in ablated lesions. T2*-GRE images inconsistently showed small hyperintense lesions that were visible adjacent to the treated joint. It has previously been shown that T2*-GRE is limited in detecting interstitial edema . Due to a trade-off between time resolution, spatial resolution, coverage, and slice thickness, we were unable to obtain meaningful perfusion results. Despite the above limitations, this study provides a basis for further development and standardization of the different techniques. A previous study indicated that the mechanism of loss of vascular integrity during MRgFUS ablation is unclear . Hynynen et al.  indicated that noninvasive vascular occlusion could be achieved using thermal MRgFUS, and Wu et al.  suggested that the cause of perfusion deficits after FUS ablation is vascular damage. Our histological study indicated that the skeletal muscle showed coagulation necrosis and damaged microvessels after FUS treatment. The damaged vessels with coagulated blood most likely played a critical role in preventing blood perfusion and delivery of MR contrast media to the core of the ablated lesion.
This study showed further that thermal ablation causes shrinkage of the cells and expansion of extracellular space. The expansion of extracellular space is unrelated to extravasation of albumin from the vascular compartment, because extracellular Gd-DTPA was unable to diffuse to the core of the lesions. A dynamic relationship exists between the extracellular and intracellular water. Water molecules exchange passively between intravascular, interstitial, and intracellular compartments under normal circumstances. Water volume in each compartment is determined by mechanisms controlling intracellular osmotic pressure, including ion homeostasis and metabolism, and water transport through sarcolemmal channels and membrane diffusion. Regulation of intracellular pH is an important element in osmotic regulation. ATP-dependent Na+/K+ pump and intracellular proteins maintain the balance between intra- and extracellular water . All these elements can be influenced by energy metabolism. It has been shown that brief myocardial ischemia leads to energy deficiency, sodium overload, cell swelling, and water dissociation from proteins because of lactate-induced acidosis, thereby increasing the intracellular fraction of free unbound water [31, 32]. We have also looked at treatment effects on bone. Although there was evidence of hemorrhage in the treated facet joint, no change in the architecture of the osteoblast was found. As these effects need more time to manifest, a longitudinal study is clearly warranted.
During FUS treatment, there should be no RF coil in the field of the sonication. Thus, the default body coil needs to be used for treatment planning but could potentially be replaced after treatment. However, we found that the default body coil of the MR system provided sufficient image quality to characterize regional tissue changes (edema, viability) after treatment. Although switching from the default body coil to a phased array coil after FUS treatment provided better SNR, it did not improve lesion detection but was time-consuming and compromised image registration.
The rise in temperature was used to define the target and may not reflect the actual temperature in the target. Furthermore, PRF-based MRI temperature mapping is very sensitive to off-resonance effects near tissue interfaces such as between the bone and muscle and may thus be inconsistent and not reliable near the facet joint. Thus, calibration is usually done within the muscle tissue, and this might be a problem in patients with severe facet arthritis where it might be very difficult to find a homogenous muscle layer. However, since lethal cell damage occurs when temperatures >55 °C are maintained for longer than a second , accurate temperature measurements are crucial to monitor the treatment and to evaluate treatment outcomes. Furthermore, post-sonication MRI could potentially be a helpful marker to identify necrotic areas where these high temperatures were met (see Fig. 8). These needs to be further investigated.
The ultimate goal of using MRgFUS for facet joint ablation is to produce lasting pain relief to the patient. Recently, Weeks et al. demonstrated the in vivo applicability of MRgFUS for facet joint palliation with relatively small energies (average 600 J) . They reported no improvements in pain scores in 40 % of the patients most likely due to the low energies used. Very recently, another group from Europe presented preliminary results from a clinical study with 35 patients where they used sonication energies of up to 2300 J . They reported no adverse events and 28 patients had a significant reduction in pain. These results suggest that there is a trade-off between safety and efficacy of MRgFUS and higher sonication energies might be more efficient. However, safety becomes increasingly a concern with higher energies, and clearly, more animal studies are warranted to investigate the optimal treatment strategy while ensuring patient safety.
In conclusion, we have evaluated several MRI pulse sequences for the assessment of FUS treatment and verified the findings with histology. We found that T2-FSE provides good image quality with positive contrast between the ablated lesion and surrounding tissues and could reduce the imaging cost and overall treatment time, as no contrast agent is needed and thus continuous FUS treatment would be possible. T1-FSE, with the aid of MR contrast media, delineated the borders of ablated lesions and provided evidence of vascular obstruction. Demonstration of perfusion deficit on first-pass perfusion imaging is challenging in the facet joints and needs further improvement. Microscopy revealed the cellular and vascular changes after ablation and provided direct evidence on the causes of hyperintensity on T2-FSE and hyperenhancement on DCE T1-FSE images. In summary, we have presented acute findings in the facet joint after FUS ablation. However, many changes on the tissue level need more time to manifest. Thus, the next important step of this research would be a longitudinal study to investigate changes over time.
The authors would like to acknowledge Dr. Jeffrey Lotz and his Orthopedic Bioengineering Laboratory for the facet joint histology as well as Dr. Lorenzo Nardo and Dr. Eugene Ozhinsky for helpful discussions.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Hynynen K, Freund WR, Cline HE, Chung AH, Watkins RD, Vetro JP, et al. A clinical, noninvasive, MR imaging-monitored ultrasound surgery method. Radiographics: a review publication of the Radiological Society of North America, Inc. 1996;16(1):185–95. doi:10.1148/radiographics.16.1.185.View ArticleGoogle Scholar
- Hectors SJ, Jacobs I, Heijman E, Keupp J, Berben M, Strijkers GJ, et al. Multiparametric MRI analysis for the evaluation of MR-guided high intensity focused ultrasound tumor treatment. NMR Biomed. 2015;28(9):1125–40. doi:10.1002/nbm.3350.View ArticlePubMedGoogle Scholar
- Hectors SJ, Jacobs I, Moonen CT, Strijkers GJ, Nicolay K. MRI methods for the evaluation of high intensity focused ultrasound tumor treatment: current status and future needs. Magn Reson Med. 2016;75(1):302–17. doi:10.1002/mrm.25758.View ArticlePubMedGoogle Scholar
- Hectors SJ, Jacobs I, Strijkers GJ, Nicolay K. Multiparametric MRI analysis for the identification of high intensity focused ultrasound-treated tumor tissue. PLoS One. 2014;9(6):e99936. doi:10.1371/journal.pone.0099936.View ArticlePubMedPubMed CentralGoogle Scholar
- Keshavarzi A, Vaezy S, Noble ML, Paun MK, Fujimoto VY. Treatment of uterine fibroid tumors in an in situ rat model using high-intensity focused ultrasound. Fertil Steril. 2003;80 Suppl 2:761–7.View ArticlePubMedGoogle Scholar
- Machtinger R, Tempany CM, Kanan Roddy A, Fennessy FM. Successful MRI-guided focused ultrasound uterine fibroid treatment despite an ostomy and significant abdominal wall scarring. ISRN obstetrics and gynecology. 2011;2011:962621. doi:10.5402/2011/962621.View ArticlePubMedPubMed CentralGoogle Scholar
- Mahmoud MZ, Alkhorayef M, Alzimami KS, Aljuhani MS, Sulieman A. High-intensity focused ultrasound (HIFU) in uterine fibroid treatment: review study. Polish journal of radiology / Polish Medical Society of Radiology. 2014;79:384–90. doi:10.12659/PJR.891110.PubMedPubMed CentralGoogle Scholar
- Stewart EA, Gedroyc WM, Tempany CM, Quade BJ, Inbar Y, Ehrenstein T, et al. Focused ultrasound treatment of uterine fibroid tumors: safety and feasibility of a noninvasive thermoablative technique. Am J Obstet Gynecol. 2003;189(1):48–54.View ArticlePubMedGoogle Scholar
- Vaezy S, Fujimoto VY, Walker C, Martin RW, Chi EY, Crum LA. Treatment of uterine fibroid tumors in a nude mouse model using high-intensity focused ultrasound. Am J Obstet Gynecol. 2000;183(1):6–11. doi:10.1067/mob.2000.105347.PubMedGoogle Scholar
- Zhang J, Feng L, Zhang B, Ren J, Li Z, Hu D, et al. Ultrasound-guided percutaneous microwave ablation for symptomatic uterine fibroid treatment—a clinical study. Int J Hyperthermia. 2011;27(5):510–6. doi:10.3109/02656736.2011.562872.View ArticlePubMedGoogle Scholar
- Catane R, Beck A, Inbar Y, Rabin T, Shabshin N, Hengst S, et al. MR-guided focused ultrasound surgery (MRgFUS) for the palliation of pain in patients with bone metastases—preliminary clinical experience. Ann Oncol. 2007;18(1):163–7. doi:10.1093/annonc/mdl335.View ArticlePubMedGoogle Scholar
- Zippel DB, Papa MZ. The use of MR imaging guided focused ultrasound in breast cancer patients: a preliminary phase one study and review. Breast Cancer. 2005;12(1):32–8.View ArticlePubMedGoogle Scholar
- Okada A, Murakami T, Mikami K, Onishi H, Tanigawa N, Marukawa T, et al. A case of hepatocellular carcinoma treated by MR-guided focused ultrasound ablation with respiratory gating. Magnetic resonance in medical sciences : MRMS : an official journal of Japan Society of Magnetic Resonance in Medicine. 2006;5(3):167–71.View ArticleGoogle Scholar
- Hynynen K, McDannold N, Clement G, Jolesz FA, Zadicario E, Killiany R, et al. Pre-clinical testing of a phased array ultrasound system for MRI-guided noninvasive surgery of the brain—a primate study. Eur J Radiol. 2006;59(2):149–56. doi:10.1016/j.ejrad.2006.04.007.View ArticlePubMedGoogle Scholar
- Vykhodtseva N, McDannold N, Hynynen K. Progress and problems in the application of focused ultrasound for blood-brain barrier disruption. Ultrasonics. 2008;48(4):279–96. doi:10.1016/j.ultras.2008.04.004.View ArticlePubMedPubMed CentralGoogle Scholar
- Harnof S, Zibly Z, Shay L, Dogadkin O, Hanannel A, Inbar Y, et al. Magnetic resonance-guided focused ultrasound treatment of facet joint pain: summary of preclinical phase. J Ther Ultrasound. 2014;2:9. doi:10.1186/2050-5736-2-9.View ArticlePubMedPubMed CentralGoogle Scholar
- Weeks EM, Platt MW, Gedroyc W. MRI-guided focused ultrasound (MRgFUS) to treat facet joint osteoarthritis low back pain—case series of an innovative new technique. Eur Radiol. 2012;22(12):2822–35. doi:10.1007/s00330-012-2628-6.View ArticlePubMedGoogle Scholar
- De Poorter J, De Wagter C, De Deene Y, Thomsen C, Stahlberg F, Achten E. Noninvasive MRI thermometry with the proton resonance frequency (PRF) method: in vivo results in human muscle. Magnetic resonance in medicine : official journal of the Society of Magnetic Resonance in Medicine / Society of Magnetic Resonance in Medicine. 1995;33(1):74–81.View ArticleGoogle Scholar
- Pauly KB, Rieke V, Holbrook AB, Grissom W, Chen J, Kaye E. MR-guidance of HIFU therapy. Conference proceedings : Annual International Conference of the IEEE Engineering in Medicine and Biology Society IEEE Engineering in Medicine and Biology Society Conference. 2009;2009:141–4. doi:10.1109/IEMBS.2009.5334680.Google Scholar
- Rieke V, Butts PK. MR thermometry. J Magn Reson Imaging. 2008;27(2):376–90. doi:10.1002/jmri.21265.View ArticlePubMedPubMed CentralGoogle Scholar
- Bazzocchi A, Napoli A, Sacconi B, Battista G, Guglielmi G, Catalano C, et al. MRI-guided focused ultrasound surgery in musculoskeletal diseases: the hot topics. Br J Radiol. 2016;89(1057):20150358. doi:10.1259/bjr.20150358.View ArticlePubMedGoogle Scholar
- Hijnen NM, Elevelt A, Grull H. Stability and trapping of magnetic resonance imaging contrast agents during high-intensity focused ultrasound ablation therapy. Invest Radiol. 2013;48(7):517–24. doi:10.1097/RLI.0b013e31829aae98.View ArticlePubMedGoogle Scholar
- Bucknor MD, Rieke V, Do L, Majumdar S, Link TM, Saeed M. MRI-guided high-intensity focused ultrasound ablation of bone: evaluation of acute findings with MR and CT imaging in a swine model. J Magn Reson Imaging. 2014;40(5):1174–80. doi:10.1002/jmri.24451.View ArticlePubMedPubMed CentralGoogle Scholar
- Fukunishi H, Funaki K, Ikuma K, Kaji Y, Sugimura K, Kitazawa R, et al. Unsuspected uterine leiomyosarcoma: magnetic resonance imaging findings before and after focused ultrasound surgery. International journal of gynecological cancer : official journal of the International Gynecological Cancer Society. 2007;17(3):724–8. doi:10.1111/j.1525-1438.2007.00818.x.View ArticleGoogle Scholar
- Kirkham AP, Emberton M, Hoh IM, Illing RO, Freeman AA, Allen C. MR imaging of prostate after treatment with high-intensity focused ultrasound. Radiology. 2008;246(3):833–44. doi:10.1148/radiol.2463062080.View ArticlePubMedGoogle Scholar
- Sung HY, Jung SE, Cho SH, Zhou K, Han JY, Han ST, et al. Long-term outcome of high-intensity focused ultrasound in advanced pancreatic cancer. Pancreas. 2011;40(7):1080–6. doi:10.1097/MPA.0b013e31821fde24.View ArticlePubMedGoogle Scholar
- Li X, Ma BC, Bolbos RI, Stahl R, Lozano J, Zuo J, et al. Quantitative assessment of bone marrow edema-like lesion and overlying cartilage in knees with osteoarthritis and anterior cruciate ligament tear using MR imaging and spectroscopic imaging at 3 tesla. J Magn Reson Imaging. 2008;28(2):453–61. doi:10.1002/jmri.21437.View ArticlePubMedPubMed CentralGoogle Scholar
- Voogt MJ, Arntz MJ, Lohle PN, Mali WP, Lampmann LE. Uterine fibroid embolisation for symptomatic uterine fibroids: a survey of clinical practice in Europe. Cardiovasc Intervent Radiol. 2011;34(4):765–73. doi:10.1007/s00270-010-9978-8.View ArticlePubMedPubMed CentralGoogle Scholar
- Wu F, Chen WZ, Bai J, Zou JZ, Wang ZL, Zhu H, et al. Tumor vessel destruction resulting from high-intensity focused ultrasound in patients with solid malignancies. Ultrasound Med Biol. 2002;28(4):535–42.View ArticlePubMedGoogle Scholar
- Hellerstein S. Fluids and electrolytes: physiology. Pediatrics in review / American Academy of Pediatrics. 1993;14(2):70–9.View ArticlePubMedGoogle Scholar
- Vandenberg JI, Rees SA, Wright AR, Powell T. Cell swelling and ion transport pathways in cardiac myocytes. Cardiovasc Res. 1996;32(1):85–97.View ArticlePubMedGoogle Scholar
- Inserte J, Garcia-Dorado D, Ruiz-Meana M, Solares J, Soler J. The role of Na+–H+ exchange occurring during hypoxia in the genesis of reoxygenation-induced myocardial oedema. J Mol Cell Cardiol. 1997;29(4):1167–75. doi:10.1006/jmcc.1996.0352.View ArticlePubMedGoogle Scholar
- Sapareto SA, Dewey WC. Thermal dose determination in cancer therapy. Int J Radiat Oncol Biol Phys. 1984;10(6):787–800.View ArticlePubMedGoogle Scholar
- Düx M. MRgFUS Frankfurt experience. 3rd ed. London, U.K: European Symposium on Focused Ultrasound Therapy; 2015.Google Scholar